Radiological Control Records
PART 6 Instrumentation and Calibration Records
761 Calibration and Operational Checks
Calibration records for fixed, portable, and laboratory radiation measuring instruments and equipment and individual
monitoring devices shall be maintained [see 835.703(d)]. These records should include frequencies, method, dates,
personnel, training, and traceability of calibration sources to National Institute of Standards and Technology or
other acceptable standards.
Calibration and maintenance records shall be maintained for instruments and equipment used for monitoring [see
835.703d]. Calibration and maintenance records should be maintained for the following equipment:
Portable survey instruments
Bioassay measurement equipment
Laboratory, counting room, and fixed radiation measuring equipme nt
Process and effluent monitors and sampling equipment
Radiation area monitors
Portal monitors and other personnel contamination monitors
Pocket and electronic dosimeters
Air sampling equipment
Tool and waste monitoring equipment
Protective clothing and equipment monitors.
Documentation of instrument operational checks shall be maintained [see 835.701(a) & 835.401(b)(4)]. Such records
should be maintained for a period not less than the calibration period of the instrument.
Maintenance results for each instrument and device shall be created and retained [see 835.703(d)]. Maintenance
histories for each instrument and device should be created and include the nature of any defects and corrective
762 Special Calibr ation Records
Records of additional tests and checks of instrumentation used in conjunction with a suspected overexposure,
questionable indication, or unusual occurrence should be retained. In addition, records of special instrument calibrations
and modifications made in accordance with Article 562.6 shall be retained [see 835.703(d)].