JOB PERFORMANCE MEASURE
30-21-150-002-004 Calibrate Honeywell ElectroniK 17 Temperature
The Honeywell Temperature Recorder must operate in accordance with plant
specifications, approved maintenance procedures and within the specifications of the
applicable Calibration Data Sheet (CDS). No personnel injuries, safety or procedure
violations. No damage to equipment or components.
Start-A-Task and Finish-A-Task; "Measuring & Test Equipment" - Course #0080;
High temperatures may exist in the immediate proximity of the Honeywell Recorder.
Exercise caution when connecting/disconnecting terminals due to restricted access at
the rear of the Honeywell Recorder. Notify Health Protection if their assistance is
Loveland 450 A Temperature Calibrator
(Category 1); Type K Thermocouple Extension Wire.
Chart Paper; Pen cartridge.
Work Clearance Permit completed, equipment/materials available, Health Protection
Operations have been notified if their assistance is required. The M&TE and other
required information asked for at the top of the CDS is filled in, and the Category 1
M&TE is verified to be within its calibration due date.